Using the full frequency resolution of a direct digital synthesizer chip outstrips the capabilities of floating point numbers. Ed takes a look at what’s needed for high-resolution frequency calibration and measurements.
By Ed Nisley
As you saw in my July article, the filter bandwidths and frequency resolution required to characterize low-frequency quartz resonators far exceeded the capabilities of my bench instruments. I decided to take a look at building a special-purpose resonator tester around a cheap direct digital synthesizer sine-wave source, because DDS generators have
advantages over traditional analog oscillators and frequency counters in computer-controlled measurement systems.
Of course, nothing is ever so simple as it seems. In this article, I’ll explain how numeric precision affects Direct Digital Synthesis (DDS) output frequency calculations, work through the effects of floating-point and fixed-point arithmetic, and show how a carefully tweaked DDS oscillator frequency varies with temperature.
You can think of a direct digital synthesizer as a lookup table holding the digitized values of an analog waveform, a counter addressing the table entries in ascending order, and a DAC converting the numbers to analog voltages. The Analog Devices AD8950 DDS chip in Photo 1 has the equivalent of a table with 232 10-bit entries defining a sine wave, a counter clocked at 125 MHz, and a differential output current-mode DAC. The PCB, complete with the DDS and a 125 MHz quartz oscillator, costs under $20 on eBay or Amazon. …
Read the full article in the September 326 issue of Circuit Cellar
Not a Subscriber yet? Become one today:
Or purchase the September 2017 issue at the CC-WebshopSponsor this Article
Circuit Cellar's editorial team comprises professional engineers, technical editors, and digital media specialists. You can reach the Editorial Department at email@example.com, @circuitcellar, and facebook.com/circuitcellar